!!! [EA-12] WAS UPGRATED TO [SAFE AQ UG], PLEASE NOTE: THE TEST STRIPS FOR SAFE AQ UG CAN’T BE USED IN EA-12.
1, Batteries are not included in package because of aviation restrictions.
2, There’re 2 device units optional–mmol/L & mg/dL, please choose it according to the standard in your country.
3, The expiry date of blood glucose test strips is up to 2022, the expiry date of uric acid test strips is up to 2021.
The sticker for coding is attached in the cover of the strip box. DO NOT discard the strip box!
Blood sample: fresh peripheral whole blood, venous whole blood Calibration sample: venous plasma Blood sample: Blood glucose-0.6μl , Uric Acid-3μl Test range: Blood glucose-20~600mg/dL, Uric Acid-3~20md/dL Test time: Blood glucose-5s, Uric Acid-25s Memory: 100 test results Battery: 2*AAA battery (No batteries in package)
1 meter + 50pcs uric acid test strips + 50pcs blood glucose test strips + 50pcs lancets
1. What is the mean of ‘E-4’ in the display? Answer: Coding Failure. Please refer to user manual
2. How to set correcting code of Uric Acid (use code chip)?
Answer: Code chip matching with test strips shall be used when you use a new vial of uric acid test strip. Find code chip in test strip package, check if the code number on code chip is in accordance with code number on test strip package. If not, you will get inaccurate test results.
If there is already a code chip in Safe AQ UG meter, please take it out and insert a new code chip.
Insert the code chip to code chip port, the meter will make a short beep sound and perform self-inspection. The meter will display code number and meter will be off after a short beep sound.
If there is an error, the meter will display E-4, Please pull out the code chip and insert again, if still failed, please contact our customer service.
Every code chip only need to be inserted once, the meter will store the code number until insert new code chip.